Rating is available when the video has been rented.
This feature is not available right now. Please try again later.
Published on Apr 24, 2014
For more information, visit http://www.bio-rad.com/yt/6/TechSuppo... This short video demonstrates troubleshooting when the Bio-Plex system returns the error message "Device has timed out", which can occur when the needle or probe cannot move to its proper location, typically due to the presence of sealing tape on the microplate, a pinched section of sample tubing, or an incorrect needle height setting.
The Bio-Plex system uses magnetic beads coupled with unique fluorpohores and customized detection antibodies for the quantitation of multiple proteins, peptides, and nucleic acids. Bio-Rad offers Bio-Plex assays for cytokines, growth factors, signal-transduction molecules, and disease biomarkers for inflammation, autoimmune disorders, cancer, diabetes and heart disease.