Rating is available when the video has been rented.
This feature is not available right now. Please try again later.
Published on Apr 24, 2014
For more information, visit http://www.bio-rad.com/yt/8/TechSuppo... This video shows how to clean the probe needle on the Bio-Plex system to keep sample pickup running smoothly and prevent needle clogging. Low bead counts are an indication that the needle needs cleaning, but it is recommended to clean it regularly before problems develop - weekly or more often with viscous or high-particulate samples.
The Bio-Plex system uses magnetic beads coupled with unique fluorpohores and customized detection antibodies for the quantitation of multiple proteins, peptides, and nucleic acids. Bio-Rad offers Bio-Plex assays for cytokines, growth factors, signal-transduction molecules, and disease biomarkers for inflammation, autoimmune disorders, cancer, diabetes and heart disease.